A. Check Door to Make Sure it is in the Locked Position and is Not Leaking.
B. Check Switch Position and Readjust if Necessary.
C. Use Level and Adjust Feet if Neccesary.
D. When Properly Filled, Water Should Cover Fill Plate.
E. Turn Clockwise to Call for More Heat.
F. Check Resistance of Element, It Should Be: * ~10 for OCM *~8 for OCR
G. Recalibrate From Cold Start or Replace and Calibrate. (PCT042)
H. Check Resistance. SHould be 1 MEG @ 77F + 20%
I. Check Resistance 20K + 10%
J. Measure Voltage From Top Terminal of NO.1 Power Switch to Common Terminal
of Function Switch. (If 120 Vac Board is Probably Open; If 0 Vac, Board
is Probably Good). This is Provided There Is No Continuity Through Function
Switch, OT Switch, and Heating Element.
K. Check Teflon Seats for Scoring or Trash Lodged Around Seats. If Scored,
Replace. If Trash or Debris, Clean and Re-Assemble.
L. Check for Activation of Switch by Main Valve Cam. Adjust Position
of Switch Arm so it is Depressed and Switch is Closed in the Sterilizer
M. Visually Check for Rocker Arm Depressing Push Rod in Sterilize Mode.
Check for Clearance and Adjust (flatten) Cam Shaft if Necessary.